提示如何在Luminex® 200™ 仪器上检查固件
演示如何为您的Luminex® 200™ 将IS2.3软件的文件转换到xPONENT® 3.1软件里
Luminex® 200™ 排除漏液故障的教学视频
This troubleshooting video informs Luminex® 200™ users on how to detect the audible alarm and fault LED on the Luminex SDS, how to troubleshoot the issue, how to drain the SDS and how to refill it.
Why should you replace the air intake filter on the XY Platform? The air intake filter keeps dust and large particles out of the XY Platform and should be replaced as part of the semi-annual maintenance routine.
Let’s take a look at the procedure for removing, cleaning, and replacing the vent filter on the Luminex 200 analyzer as part of the semi-annual maintenance schedule.
In this video, we will go over how to clean the exterior surfaces of the Luminex 200. This routine is part of the monthly maintenance schedule to keep the Luminex 200 clean and sterile.
Let’s take a look at the procedure of performing a visual inspection of the Luminex 200 Instrument. This routine is part of the weekly maintenance schedule to keep the system clean and to help identify potential issues.
Replacing the seal will prevent leaks or pressure loss in the fluidics system. This preventative maintenance procedure is to be used if the seal is not performing properly.
The intake air filter needs to be replaced every six months to keep the fluidics bay sterile and clean of particles that build up on the filter over time.
Following the startup and shutdown daily maintenance routines help ensure accurate test results by properly cleaning and maintaining the instrument.
Keeping the sheath filter clean helps ensure accurate test results by keeping the fluidics system clean of particles that buildup on the filter. The filter needs to be replaced annually.
Let’s take a look at the procedure for removing and replacing the fuses on the Luminex 200. This process is only required if a fuse is tripped.
Flushing the system ensures that all the fluidics lines and the chamber in the Luminex 200 are clean and debris free. It is important to flush the system once per week.
视频演示如何从IS2.3 升级到xPONENT® 3.1
对Luminex® 200™ 进行月保养或者故障排除时，应该校准和验证以确定仪器正常