This Quick Tip shows the user how to check firmware on the Luminex® 200™ Instrument.
This Quick Tip indicates how to import the License Key into xPONENT® for the Luminex® 200™ Instrument.
This Quick Tip indicates how to convert files from IS 2.3 to xPONENT® 3.1 for your Luminex® 200™ Instrument.
This troubleshooting video informs Luminex® 200™ users on how to detect the audible alarm and fault LED on the Luminex SDS, how to troubleshoot the issue, how to drain the SDS and how to refill it.
This tutorial will help troubleshoot a fluid leak issue on the Luminex® 200™ Instrument.
This video will provide you with troubleshooting steps if you have a Luminex 200 Instrument with acquisition issues. Let our Technical Support representative guide you through these steps in order to get your instrument working again.
The steps listed in this brief video will walk you through how to adjust the probe height on your Luminex® 200™ instrument.
This video will demonstrate the steps to take bypass the sample probe to check if there are issues with the probe.
This tutorial will teach you how to run CON1 as an assay on your Luminex® 200™ instrument.
This video demonstrates how to upgrade IS 2.3 software to xPONENT® 3.1.
The stringent clean process consists of four stages that can solve various issues on the Luminex 200 such as Calibration or control failures, probe or instrument clogs, and more.
There are several reasons why the probe may need to be cleaned or the height adjusted on your Luminex 200.
The process of calibration and verification of your Luminex 200 instrument should be done as part of your monthly maintenance for your instrument or for troubleshooting purposes.